- I want to check that my chip is reading properly. This is the heart of the RheoSense viscometer. We recommend recalibration every six months. To do checks, recommend using NIST traceable standards because these standards are formally certified by NIST and have traceability. It is important to be working with reliable standards.
For this measurement, I am using an N10 Oil. Below is the specific certificate of the bottle I used:
- With the N10 standard, I loaded the disposable pipette with the N10 and ran my meausurements. Below are the results I got!
Now you can run in what the instrument is programmed as 'AUTO' mode. What this means is that the instrument will run at 50% of the chip capabilities. This is extremely handy when you are doing quick measurements or working with unknown samples.
When you want to run the measurement at a particular shear rate, you can choose 'ADVANCED' mode and plug in the shear rate or flow rate you would like to run. An example would be run #7 (see below), where the shear rate is 200 1/s vs. all of the other runs that were done at 774.9, 774.8 1/s.
2. Based on the certified standard, at 25 °C, viscosity should be 14.94 cP. With all five runs, you can see that the largest % difference is at about 0.87%.
See PDF versions of:
Contact us for more information on microVISC™ Viscometer! We also carry NIST traceable oil and water based standards!